Temperature Data Processing
Water Depth: 2864.4 mbrf
Temperature Tool Used: LDEO-TLT
Depth versus time recording available: NO
Logging string 1: ACT/GST/NGT
Wireline heave compensator was used to counter ship heave resulting from the mild sea conditions (0.3-1.25 m).
The LDEO-TLT tool is a self-contained, high precision, low-temperature logging tool that is attached to the bottom of the Schlumberger tool strings. The tool provides two temperature measurements (in degree Celsius, recorded by a fast-response and a slow-response thermistor. The fast-response thermistor, though low in accuracy, is able to detect sudden, small temperature excursions caused by fluid flow between the formation and the borehole. The highly accurate, slow-response thermistor can be used to estimate heat flow. Pressure and the two temperature measurements are recorded as a function of time: conversion to depth can be based on the pressure reading (Legs 123-157) or, preferably, on simultaneous recording (by Schlumberger) of depth and time (Legs 159-181).
No original data was available at the time of reprocessing, in June 2000. Original processing was performed in 1995. The pressure data was smoothed with a 10-point running average and then decimated by taking every 5th point. The temperature data was smoothed with a 5-point running average and then decimated by taking every 5th point. Due to tool malfunction, there are no temperature readings from the fast thermistor. A linear relationship of pressure versus depth had been calculated from the pressure reading at the mudline and the maximum logged depth. The pressure at the mudline corresponds to the pressure recorded by the tool during the calibration stop (about 5 minutes), which takes place at the mudline on each logging run. The recording stopped at 434 mbsf during the uplog.
The procedure described above does not fully account for the vagaries of the pressure readings that may result in lots of ups and downs in the generated depth channel. Further problems arise because of pumping during logging operations, which affects the pressure, especially when the Side Entry Sub is used. Also, whenever heavy pills of mud are used, the pressure-depth calculation is affected, resulting in a non-linear effect that is difficult to account for. If the pressure conversion coefficient is recalculated for the mudline, the resultant total depth is often wrong.
Information about the temperature logging operations can be found in the Site Chapter (Operations, Downhole Measurements, and Heat Flow sections), ODP IR volume 157.
For further information about the processing, please contact:
E-mail: Cristina Broglia